

The Agilent B1510A High power source/monitor unit (HPSMU) features range up to 200 V/1 A with 4quadrant operation. Minimum measurement resolution 10 fA/2 μV. It uses two slots in a B1500A Mainframe. Specifications.
Voltage (high resolution ADC). Range, Force resolution, Measure resolution, Force accuracy ±(% read value + offset voltage V), Measure accuracy (% read value + offset voltage V), Maximum current . ±2 V, 100 μV, 2 μV, ±(0.018% + 400 μV), ±(0.01% + 140 μV), 1 A. ±20 V, 1 mV, 20 μV, ±(0.018% + 3 mV), ±(0.009% + 900 μV), 1 A. ±40 V, 2 mV, 40 μV, ±(0.018% + 6 mV), ±(0.01% + 1 mV), 500 mA. ±100 V, 5 mV, 100 μV, ±(0.018% + 15 mV), ±(0.012% + 2.5 mV), 125 mA. ±200 V, 10 mV, 200 μV, ±(0.018% + 30 mV), ±(0.014% + 2.8 mV), 50 mA.
Current (high resolution ADC). Range, Force resolution, Measure resolution, Force accuracy ±(% + A + A), Measure accuracy (% + A + A), Max voltage. ±1 nA, 50 fA, 10 fA, ±(0.1%+300 fA+1 fA x Vo), ±(0.1%+200 fA+1 fA x Vo), 200 V. ±10 nA, 500 fA, 10 fA, ±(0.1%+3 pA+10 fA x Vo), ±(0.1%+1 pA+10 fA x Vo), 200 V. ±100 nA, 5 pA, 100 fA, ±(0.05%+30 pA+100 fA x Vo), ±(0.05%+20 pA+100 fA x Vo), 200 V. ±1 μA, 50 pA, 1 pA, ±(0.05%+300 pA+1 pA x Vo), ±(0.05%+100 pA+1 pA x Vo), 200 V. ±10 μA, 500 pA, 10 pA, ±(0.05%+3 nA+10 pA x Vo), ±(0.04%+2 nA+10 pA x Vo), 200 V. ±100 μA, 5 nA, 100 pA, ±(0.035%+15 nA+100 pA x Vo), ±(0.03%+3 nA+100 pA x Vo), 200 V. ±1 mA, 50 nA, 1 nA, ±(0.04%+150 nA+1 nA x Vo), ±(0.03%+60 nA+1 nA x Vo), 200 V. ±10 mA, 500 nA, 10 nA, ±(0.04%+1.5 μA+10 nA x Vo), ±(0.03%+200 nA+10 nA x Vo), 200 V. ±100 mA, 5 μA, 100 nA, ±(0.045%+15 μA+100 nA x Vo), ±(0.04%+6 μA+100 nA x Vo), 200 V (Io ≤ 50 mA)/ 100 V (50 mA < Io ≤ 125 mA)/ 40 V (125 mA < Io ≤ 500 mA)/ 20 V (500 mA < Io ≤ 1 A). ±1 A, 50 μA, 1 μA, ±(0.4%+300 μA+1 μA x Vo), ±(0.4%+150 μA+1 μA x Vo), 200 V (Io ≤ 50 mA)/ 100 V (50 mA < Io ≤ 125 mA)/ 40 V (125 mA < Io ≤ 500 mA)/ 20 V (500 mA < Io ≤ 1 A).
Power consumption. Voltage source mode (Ic is the current compliance setting): 20 x Ic (W) (2 to 20V), 40 x Ic (W) (40V), 100 x Ic (W) (100V), 200 x Ic (W) (200V). Current source mode (Vc is the voltage compliance setting and Io is output current): 20 x Io (W) (Vc ≤ 20), 40 x Io (W) (20 < Vc ≤ 40), 100 x Io (W) (40 < Vc ≤ 100), 200 x Io (W) (100 < Vc ≤ 200).
